Background: Cardiorespiratory fitness (CRF), an important index of physical fitness, is the ability to inhale and provide oxygen to the exercising muscle. However, despite its importance, the current gold standard for measuring CRF is impractical, requiring maximal exercise from the participants.
Objective: This study aimed to develop a convenient and practical estimation model for CRF using data collected from daily life with a wristwatch-type device.
Methods: A total of 191 subjects, aged 20 to 65 years, participated in this study. Maximal oxygen uptake (VO2 max), a standard measure of CRF, was measured with a maximal exercise test. Heart rate (HR) and physical activity data were collected using a commercial wristwatch-type fitness tracker (Fitbit; Fitbit Charge; Fitbit) for 3 consecutive days. Maximal activity energy expenditure (aEEmax) and slope between HR and physical activity were calculated using a linear regression. A VO2 max estimation model was built using multiple linear regression with data on age, sex, height, percent body fat, aEEmax, and the slope. The result was validated with 2 different cross-validation methods.
Results: aEEmax showed a moderate correlation with VO2 max (r=0.50). The correlation coefficient for the multiple linear regression model was 0.81, and the SE of estimate (SEE) was 3.518 mL/kg/min. The regression model was cross-validated through the predicted residual error sum of square (PRESS). The PRESS correlation coefficient was 0.79, and the PRESS SEE was 3.667 mL/kg/min. The model was further validated by dividing it into different subgroups and calculating the constant error (CE) where a low CE showed that the model does not significantly overestimate or underestimate VO2 max.
Conclusions: This study proposes a CRF estimation method using data collected by a wristwatch-type fitness tracker without any specific protocol for a wide range of the population.
Cardiorespiratory fitness (CRF) is an important component of physical fitness, representing the body’s ability to take oxygen in and deliver this oxygen to muscle cells throughout the body during physical activity. Previous studies have emphasized the importance of CRF, providing convincing evidence that CRF is closely related to all-cause mortality [, ]. In addition, CRF is known to be correlated with various physiological factors, such as body composition and blood pressure, and psychological factors, such as depression [ - ]. Erikssen et al [ ] have reported that a change in physical fitness is a strong predictor of mortality. They found that a small improvement in physical fitness can significantly lower the risk of death.
Maximal oxygen uptake (VO2 max) is regarded as a representative feature of CRF. The current gold standard for measuring VO2 max is a metabolic gas analysis during a maximal graded exercise test (GXT) on a treadmill or other equipment, such as cycle ergometer. Even though the maximal exercise test provides an accurate measurement of VO2 max, there are several limitations. The maximal exercise requires a high level of motivation from the subject and should be performed under medical supervision for older or high-risk subjects who need this test the most . Furthermore, the gas analysis requires expensive equipment and a trained technician to operate the process [ ]. In addition, because of the high cost and inconvenience, it is impractical to repeat the maximal exercise test to regularly monitor VO2 max.
Several estimation models have been developed to estimate VO2 max. Some of these models have developed a submaximal exercise protocol in an attempt to overcome the limitation of the maximal test [- ]. Submaximal models obtain exercise-related data through a specified exercise protocol, such as shuttle run, and build estimation models along with other anthropometric features. Although submaximal models have overcome some of the limitations of the GXT, they still require trained personnel to conduct the submaximal test, and familiarity with the exercise protocol could affect the results of the test [ ], making it unsuitable for regular VO2 max monitoring. There are other estimation models that do not involve an exercise protocol [ , ]. These models estimate VO2 max by collecting data from physical activity and heart rate (HR) from daily life and using the relationship between the collected data and VO2 max. Although these methods are more suitable for regular VO2 max measurement, they are time-consuming (requiring a week of data collection) and use multiple devices, making it uncomfortable for application in daily life.
In our previous study , we developed a nonexercise VO2 max estimation model using a new feature, maximum activity energy expenditure (aEEmax), which was calculated using activity energy expenditure and HR. Using aEEmax, we were able to build an accurate estimation model. However, aEEmax and our previous model were validated only in homogenous subjects, young Asian males. Furthermore, the device was worn on the chest, which might cause discomfort when used in daily life.
The aim of this study was to overcome the limitations of our previous study by using a wristwatch-type fitness tracker with various groups in terms of age and sex. We also sought to develop a new VO2 max estimation model using aEEmax and the slope between physical activity and HR as new features, which could be applied to daily life data collected from a single convenient device worn on a wrist with a relatively short estimation time.
A total of 240 participants were recruited for this study. All participants completed the Physical Activity Readiness Questionnaire and health evaluation, including medical history related to cardiovascular disease, hypertension, and/or diabetes. Only participants without such medical history were included for this study. There were a total of 6 groups, divided according to age (20 to 35, 36 to 50, and 51 to 65 years) and sex, and there were 40 subjects for each group. Subjects who failed to achieve VO2 max were excluded from the study. The achievement of VO2 max was defined by accomplishing at least 2 of the following 3 criteria: a respiratory exchange ratio reaching >1.2, plateau of VO2 despite increasing work load, or self-reported volitional fatigue . Participants who did not wear the device for 3 days or participants with data loss were also excluded. A total of 49 subjects were excluded because of failure to achieve VO2 max or unappropriated data collection. The characteristics of the participants are shown in .
|Characteristics||Male, mean (SD)||Female, mean (SD)|
|20-35 years (n=34)||36-50 years (n=26)||51-65 years (n=30)||20-35 years (n=36)||36-50 years (n=35)||51-65 years (n=30)|
|Height (cm)||174.3 (5.6)||172.6 (6.1)||167.1 (4.9)||161.9 (5.4)||160.0 (5.2)||155.3 (4.8)|
|Weight (kg)||73.9 (8.0)||74.7 (9.7)||67.0 (6.1)||55.8 (7.3)||60.6 (6.1)||56.2 (6.1)|
|Percent body fat||20.4 (5.2)||24.8 (4.9)||24.2 (5.1)||29.4 (6.7)||33.9 (4.9)||33.8 (5.4)|
|aEEmaxa (kcal/kg/h)||141.0 (14.6)||123.7 (15.1)||111.2 (11.0)||112.5 (15.8)||107.7 (12.1)||102.0 (10.3)|
|Slope (kcal/kg/h/bpm)||1.10 (0.14)||1.04 (0.13)||1.04 (0.12)||0.92 (0.17)||0.98 (0.14)||0.96 (0.13)|
|VO2 maxb (mL/kg/min)||42.3 (3.6)||39.9 (3.5)||38.1 (4.6)||35.3 (3.5)||31.4 (4.1)||30.5 (3.9)|
aaEEmax: maximal activity energy expenditure.
bVO2 max: maximal oxygen uptake.
Body mass and height were measured using a medical scale with a stadiometer (BSM330; InBody). Body mass was measured to the nearest 0.1 kg, and height was measured to the nearest 0.1 cm. Percent body fat was measured using a bioimpedance analysis (InBody720; InBody) to the nearest 0.1%.
Measurement of Maximal Oxygen Uptake
The reference VO2 max value was measured using the modified Bruce protocol. The equipment used in the modified Bruce protocol includes a respiration gas analyzer (Vmax Encore System; CareFusion) and an aerobic exercise test system (CASE v6.61; GE Healthcare). Standard 12-lead electrocardiogram (ECG), blood oxygen saturation, and blood pressure were measured throughout the procedure.
Before performing the modified Bruce protocol, the baseline physiologic measures for all devices used were measured in a resting state for 5 min and subsequently in a standing position. The modified Bruce protocol was performed immediately after the baseline measurement. The treadmill’s velocity and slope increased at 3-min intervals until the subject reached VO2 max.
Participants wore a Fitbit (Fitbit Charge; Fitbit) on the left wrist for 3 consecutive days. From our previous study, we have shown that a minimum of 15 hours of physical activity data are required to acquire aEEmax . In total, 3 consecutive days, regardless of weekday or weekend days, were shown to be enough to obtain the data needed based on the study. The Fitbit simultaneously measured HR and daily physical activity in terms of metabolic equivalent (kcal/kg/h), which was an expression of energy expenditure of activities.
The participants removed the sensor during sleep or showering. Data obtained from the sensor were retrieved after the experiment via the internet. The Fitbit returned HR and metabolic equivalent data for every 1 min.
Moving average filter was applied to both HR and physical activity data. After filtering the data, only data points at which both HR and physical activity data increased were selected as a period of physical activity for further processing. This was done by differentiating the data and selecting where both differentiated data were positive.
shows the HR (upper graph) and physical activity (lower graph) for a representative participant’s filtered data over time. The shaded area under the physical activity curve represents the periods of increasing HR and physical activity. The scatter plot for HR versus physical activity is shown in . A simple linear regression was performed between HR and physical activity to estimate aEEmax. The part of data where HR was greater than 120 beats/min was selected to remove the data in which the relationship between HR and physical activity was nonlinear [ ]. The maximal HR was calculated as 200–age×0.67 for women and 216–age×0.93 for men [ ]. The physical activity value of the point of intersection between the maximum HR and the regression line was defined as the aEEmax. The slope of the regression line was also used as a feature to estimate VO2 max; hereafter, it will be referred as the slope. After the calculation of aEEmax and the slope, a multiple linear regression model was developed with aEEmax and anthropometric values to estimate VO2 max.
Pearson correlation coefficient was calculated between the independent variables (age, percent body fat, height, gender, aEEmax, and slope) and the measured VO2 max. The regression model for estimating VO2 max was evaluated with the coefficients of determination (adjusted R2) and absolute SE of the estimate (SEE). The predicted residual error sum of squares (PRESS) statistic method was selected for cross-validation of the model . The PRESS statistic is a cross-validation method calculating the error for each case by excluding a case each time from generating the estimation model and applying the model to the excluded case. The PRESS adjusted R2 (R2p) and the PRESS SEE (SEEp) were calculated as 1 – (PRESS/SS) and . The model was further validated by dividing it into different subgroups and calculating the constant error (CE) for each group. The standard for recruiting participants was to retain diversity. However, we wanted to observe CE based on age, sex, and VO2max level. The median value for age and VO2max was chosen to divide the groups. All signal processing, cross-validation, and statistical analyses were performed using MATLAB (MATLAB2017a; MathWorks).
This study protocol was reviewed and approved by the Institutional Review Board of the Seoul National University Hospital (IRB No. 1505-022-669). Written informed consent was submitted by all subjects when they were enrolled. This study followed the Helsinki Declaration.
The general characteristics of all subjects are summarized in. The average value for age, weight, height, and percent body fat for excluded male subjects were 42.0, 72.8 kg, 171.0 cm, and 25.0% respectively. For excluded female subjects, the average values were 41.7, 56.5 kg, 158.4 cm, and 31.6%, respectively. A student t test was performed to compare the P value for age, weight, height, and percent body fat between included and excluded subjects. For male subjects, the P values were .646, .937, .883, and .026, respectively. For female subjects, the P values were .936, .415, .663, and .497 respectively.
The Pearson correlations between VO2 max and selected features are shown in. The Pearson correlations between selected features are also summarized. The correlations between VO2 max and independent variables were all statistically significant (P<.001 for all). The independent variable that showed the highest correlation was sex, with a correlation of .675. The lowest correlation for the model was the slope, with a correlation of .237.
The multiple linear regression analysis for the model is shown in. The scatter plot for measured VO2 max versus predicted VO2 max is shown in . The R2 for the model was 0.651, and the SEE was 3.518 mL/kg/min. As shown in , the decrease in R2 and the increase in SEE were small for the cross-validation result of the PRESS method. R2 decreased by 0.032 and SEE increased by 0.148 mL/kg/min. The scatter plot of the multiple linear regression for the model is shown in .
The model was further validated by dividing the groups into various subgroups and calculating the CE and SD. Each subgroup was divided into 2 groups according to age, sex, and measured VO2 max. The results are shown in. The CEs were positive for the younger group and negative for the older groups. The CE for the younger group was 0.024 mL/kg/min. The CE for the older group was −0.021 mL/kg/min. The CEs were all positive for the subgroups divided according to sex. The CE was positive for individuals with high VO2 max and negative for individuals with low VO2 max. However, as shown in , CE values were low for all subgroups, indicating that our model does not overestimate or underestimate VO2 max.
|Independent variables||VO2 maxa||Age||Height||Sex||Percent fat||aEEmaxb|
aVO2 max: maximal oxygen uptake.
baEEmax: maximal activity energy expenditure.
|Independent variables||Fitbit model|
|Percent body fat||−0.242||−.296|
aBeta is the normalized coefficient of the model.
b—: not applicable.
caEEmax: maximal activity energy expenditure.
dR is the Pearson correlation.
eSEE: SE of estimate.
fRP and SEEP are the cross-validated results of the model.
|High VO2 maxb||77 (40.3)||1.535||3.565|
|Low VO2 max||114 (59.7)||−1.488||3.232|
aCE: constant error.
bVO2 max: maximal oxygen uptake.
In this study, we developed VO2 max estimation models with data collected from the commercially available device, Fitbit, which could estimate VO2 max conveniently from daily life. The commercial device used in this study provided data on physical activity, along with its physiological response (HR). Even though some previous studies have reported that Fitbit does not always generate accurate data [, ], our method for estimating VO2 max does not depend on the absolute value of each data point. It rather depends on the trend of a large set of data points and would balance the inaccuracy of a single data point. Even though Fitbit data were biased under certain conditions, a large number of data collected from daily life would minimize this bias. This characteristic of our 2 novel features would only need calibration to be applied with other hardware and processing methods. Previous studies reported that lower HR was not linearly correlated with physical activity [ ]. It is also known that heterogeneous recovery of HR after physical exercise does not have a clear correlation with VO2 max [ ]. Therefore, in an attempt to select the data during physical activity, we have selected periods of data where HR is greater than 120 beats/min and where both HR and physical activity increased at the same time to calculate aEEmax and the slope. By calculating 2 features from the linear relationship between HR and energy expenditure, our model would estimate VO2max accurately without any protocol or training.
In our previous study , aEEmax and our estimation model were validated with a homogenous group of young healthy Asian males using aEEmax and BMI (VO2max=0.192 x aEEmax – 0.708 x BMI). In this study, we have validated aEEmax and the model with a large pool of subjects, including both men and women aged from 20 to 65 years. The validation of the model was performed by cross-validation with the PRESS method and by calculating the CE of the subgroups. The cross-validated result shows that the validation sample fitted well with the model with little error. Furthermore, the CE for the model shows that our methodology did not significantly overestimate nor underestimate VO2 max for all subgroups, whereas other studies [ , ] have reported significant overestimation and underestimation with both highly and poorly fitted individuals.
To the best of our knowledge, this study is the first to provide a VO2 max estimation model with a commercially available device on a wrist without any specific protocol. Tönis et al used a submaximal exercise protocol to estimate VO2 max . Polar Electro Oy Inc developed a nonexercise protocol, Polar Fitness Test, and devices to estimate VO2 max. However, Esco et al [ ] reported that the Polar Fitness Test had low accuracy for estimating VO2 max when it was tested with one of its own products. Altini et al [ ] developed a nonexercise estimation model for VO2 max with data collected from daily life; however, they used an ECG necklace with a wet electrode attached to the chest and stomach, which could cause an inconvenience when used in daily life. Our method allows individuals to measure their VO2 max on a wrist without the requirement for any electrode attachment. In addition, although other smartwatches, including Fitbit, require a specific protocol, such as running for at least 10 min on flat terrain, our protocol does not require any protocol and allows for easy monitoring of physical fitness.
Instead of using the absolute value of HR or physical activity data, we developed 2 new features to portray physical fitness. The change in HR for a given physical activity differs depending on the physical fitness of a subject . Thus, the slope between HR and physical activity would be smaller and aEEmax would be larger for a subject with higher VO2 max. As shown in , aEEmax has a moderate correlation with VO2 max, supporting our hypothesis. Features based on physical activity can easily fluctuate depending on change in the short-term lifestyle of the subject during the period of data collection. Those features would be vulnerable to a sudden increase or decrease in the amount of physical activity. On the contrary, aEEmax and the slope represent the relationship between physical activity and HR and thus would be less affected by a sudden short-term change in physical activity. In addition, aEEmax and slope have been shown to be applicable to a wide range of subject ages and different genders. Other studies [ - ] have been validated with a relatively homogenous group compared with this study. The percent body fat used in this study was obtained using a professional bioimpedance analyzer. However, we have used general percent body fat which did not necessarily require a professional analyzer. There are products available, such as an AURA device, and ongoing studies about measuring body fat percentage from the wrist [ ]. These efforts will make the measurement of percent body fat more accessible to the public.
There are limitations to this study. First, our methodology needs to be validated with more devices worn on a wrist. There are many commercially available devices that provide physical activity and HR data. To provide a more generalized VO2 max estimation method, it is important to prove device independency of our method. In addition, this study was conducted with healthy subjects who were not taking any medication that might affect the HR. Another limitation of this study was error with maximal HR calculated from a basic population-derived formula. A more accurate method for calculating maximal HR could increase the accuracy of our model. A future study could include subjects who are on cardiac-related medication. Additionally, in our previous study, we have shown that 900 min of data were enough to calculate aEEmax; however, it would be worthwhile to observe change in the correlation coefficient of aEEmax for a longer period of time.
In summary, we have developed a new estimation model for VO2 max using novel features, aEEmax and the slope between physical activity and HR, along with other anthropometric variables. The new features represent the relationship between physical activity and its physiological response. The high correlation between VO2 max and aEEmax is in agreement with our previous study and supports our hypothesis. Our model requires data from only 3 days of daily life, without any specific exercise protocol. This hypothesis was validated with a diverse and large number of participants based on age and sex. Furthermore, all material required for our study is available in the conventional market as fully built products. The result of this study allows individuals to measure their VO2 max conveniently in their daily life without any burden of an exercise protocol and allows them to easily monitor physical fitness.
This study was supported by the Samsung Research Funding Center of Samsung Electronics under Project Number SRFC-IT1402-04. This study was also supported by the Bio & Medical Technology Development Program of the National Research Foundation (NRF), funded by the Korean government (Ministry of Science, ICT, and Future Planning [MSIP]; No. 2016M3A9F1939646). The authors would like to thank all participants.
Conflicts of Interest
- Blair SN, Kampert JB, Kohl HW, Barlow CE, Macera CA, Paffenbarger RS, et al. Influences of cardiorespiratory fitness and other precursors on cardiovascular disease and all-cause mortality in men and women. J Am Med Assoc 1996 Jul 17;276(3):205-210. [Medline]
- Lee DC, Artero EG, Sui X, Blair SN. Mortality trends in the general population: the importance of cardiorespiratory fitness. J Psychopharmacol 2010 Nov;24(4 Suppl):27-35 [FREE Full text] [CrossRef] [Medline]
- Gerber M, Lindwall M, Lindegård A, Börjesson M, Jonsdottir IH. Cardiorespiratory fitness protects against stress-related symptoms of burnout and depression. Patient Educ Couns 2013 Oct;93(1):146-152. [CrossRef] [Medline]
- Kaminsky LA, Arena R, Beckie TM, Brubaker PH, Church TS, Forman DE, et al. The importance of cardiorespiratory fitness in the United States: the need for a national registry: a policy statement from the American Heart Association. Circulation 2013;127(5):652-662. [Medline]
- Erikssen G, Liestøl K, Bjørnholt J, Thaulow E, Sandvik L, Erikssen J. Changes in physical fitness and changes in mortality. Lancet 1998 Sep 05;352(9130):759-762. [CrossRef] [Medline]
- Akay MF, Inan C, Bradshaw DI, George JD. Support vector regression and multilayer feed forward neural networks for non-exercise prediction of VO2max. Expert Syst Appl 2009 Aug;36(6):10112-10119. [CrossRef]
- Ernesto C, Martins da Silva F, Pereira L, de Melo G. Cross Validation of Different Equations to Predict Aerobic Fitness by the Shuttle Run 20 Meters Test in Brazilian Students. Journal of Exercise Physiology Online 2015;18(1).
- Ekblom-Bak E, Björkman F, Hellenius ML, Ekblom B. A new submaximal cycle ergometer test for prediction of VO2max. Scand J Med Sci Sports 2014 Apr;24(2):319-326. [CrossRef] [Medline]
- Larsen GE, George JD, Alexander JL, Fellingham GW, Aldana SG, Parcell AC. Prediction of maximum oxygen consumption from walking, jogging, or running. Res Q Exerc Sport 2002 Mar;73(1):66-72. [CrossRef] [Medline]
- Matabuena M, Vidal JC, Hayes PR, Huelin Trillo F. A 6-minute sub-maximal run test to predict VO max. J Sports Sci 2018 Nov;36(22):2531-2536. [CrossRef] [Medline]
- Noonan V, Dean E. Submaximal exercise testing: clinical application and interpretation. Phys Ther 2000 Aug;80(8):782-807. [Medline]
- Cao ZB, Miyatake N, Higuchi M, Miyachi M, Ishikawa-Takata K, Tabata I. Predicting VO2max with an objectively measured physical activity in Japanese women. Med Sci Sports Exerc 2010 Jan;42(1):179-186. [CrossRef] [Medline]
- Plasqui G, Westerterp KR. Accelerometry and heart rate as a measure of physical fitness: proof of concept. Med Sci Sports Exerc 2005 May;37(5):872-876. [Medline]
- Ahn JW, Hwang SH, Yoon C, Lee J, Kim HC, Yoon HJ. Unobtrusive estimation of cardiorespiratory fitness with daily activity in healthy young men. J Korean Med Sci 2017 Dec;32(12):1947-1952 [FREE Full text] [CrossRef] [Medline]
- Midgley AW, McNaughton LR, Polman R, Marchant D. Criteria for determination of maximal oxygen uptake: a brief critique and recommendations for future research. Sports Med 2007;37(12):1019-1028. [CrossRef] [Medline]
- Ainslie PN, Reilly T, Westerterp KR. Estimating human energy expenditure: a review of techniques with particular reference to doubly labelled water. Sports Med 2003;33(9):683-698. [CrossRef] [Medline]
- Casteel B. American college of cardiology. 2014. The Heart Responds Differently to Exercise in Men vs Women URL: https://www.acc.org/about-acc/press-releases/2014/03/27/12/29/allison-peak-hr-pr [accessed 2019-06-03] [WebCite Cache]
- Holiday DB, Ballard JE, McKeown BC. PRESS-related statistics: regression tools for cross-validation and case diagnostics. Med Sci Sports Exerc 1995 Apr;27(4):612-620. [Medline]
- Feehan LM, Geldman J, Sayre EC, Park C, Ezzat AM, Yoo JY, et al. Accuracy of Fitbit devices: systematic review and narrative syntheses of quantitative data. JMIR Mhealth Uhealth 2018 Aug 09;6(8):e10527 [FREE Full text] [CrossRef] [Medline]
- Wang R, Blackburn G, Desai M, Phelan D, Gillinov L, Houghtaling P, et al. Accuracy of wrist-worn heart rate monitors. JAMA Cardiol 2017 Dec 01;2(1):104-106. [CrossRef] [Medline]
- Sartor F, Vernillo G, de Morree HM, Bonomi AG, La Torre A, Kubis HP, et al. Estimation of maximal oxygen uptake via submaximal exercise testing in sports, clinical, and home settings. Sports Med 2013 Sep;43(9):865-873. [CrossRef] [Medline]
- Jackson AS, Blair SN, Mahar MT, Wier LT, Ross RM, Stuteville JE. Prediction of functional aerobic capacity without exercise testing. Med Sci Sports Exerc 1990 Dec;22(6):863-870. [Medline]
- Tönis TM, Gorter K, Vollenbroek-Hutten MM, Hermens H. Comparing VO2max determined by using the relation between heart rate and accelerometry with submaximal estimated VO2max. J Sports Med Phys Fitness 2012 Aug;52(4):337-343. [Medline]
- Astorino T, Baker J, Brock S, Dalleck L, Goulet E, Gotshall R, et al. Cross-Validation of the Polar Fitness Test TM via the Polar F11 Heart Rate Monitor in Predicting VO 2 Max. J Exerc Physiol 2011;14(5) [FREE Full text]
- Altini M, Casale P, Penders J, Amft O. Cardiorespiratory fitness estimation in free-living using wearable sensors. Artif Intell Med 2016 Mar;68:37-46. [CrossRef] [Medline]
- Fudge BW, Wilson J, Easton C, Irwin L, Clark J, Haddow O, et al. Estimation of oxygen uptake during fast running using accelerometry and heart rate. Med Sci Sports Exerc 2007 Jan;39(1):192-198. [CrossRef] [Medline]
- Malek MH, Housh TJ, Berger DE, Coburn JW, Beck TW. A new nonexercise-based VO2(max) equation for aerobically trained females. Med Sci Sports Exerc 2004 Oct;36(10):1804-1810. [Medline]
- Bradshaw DI, George JD, Hyde A, LaMonte MJ, Vehrs PR, Hager RL, et al. An accurate VO2max nonexercise regression model for 18-65-year-old adults. Res Q Exerc Sport 2005 Dec;76(4):426-432. [CrossRef] [Medline]
- Jung M, Namkoong K, Lee Y, Koh YJ, Eom K, Jang H, et al. Wrist-wearable bioelectrical impedance analyzer with contact resistance compensation function. IEEE Sensors 2016:1-3. [CrossRef]
|aEEmax: maximal activity energy expenditure|
|CE: constant error|
|CRF: cardiorespiratory fitness|
|GXT: graded exercise test|
|HR: heart rate|
|PRESS: predicted residual error sum of square|
|SEE: SE of estimate|
|VO2max: maximal oxygen uptake|
Edited by G Eysenbach; submitted 08.01.19; peer-reviewed by G Signorelli, M Altini, K Lu; comments to author 28.03.19; revised version received 22.04.19; accepted 17.05.19; published 13.06.19Copyright
©Soon Bin Kwon, Joong Woo Ahn, Seung Min Lee, Joonnyong Lee, Dongheon Lee, Jeeyoung Hong, Hee Chan Kim, Hyung-Jin Yoon. Originally published in JMIR Mhealth and Uhealth (http://mhealth.jmir.org), 13.06.2019.
This is an open-access article distributed under the terms of the Creative Commons Attribution License (https://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work, first published in JMIR mhealth and uhealth, is properly cited. The complete bibliographic information, a link to the original publication on http://mhealth.jmir.org/, as well as this copyright and license information must be included.